The image below shows the Vicam attached to the microscope eyepiece
as it would be in use.
This image shows the adapter on the microscope with the Vicam.
The image below is a little better view of the adapter. As you can see, it is extremely primitive but it does work quite well. When making one it is important to cut away just enough of the pipe to allow the Vicam to be centered over the tube and, therefore, centered over the eyepiece. The distance between the cuff that fits over the eyepiece and the screw should be long enough to allow some wiggle room between the camera lens and the eyepiece. No part of the camera or adapter should ever touch the eyepiece lens.
Click here to see some results.